Fixed Mount Stroboscopes
LS-FCM Flash Control Module
When manual flash rate adjustment is needed, the LS-FCM offer a convenient low-cost solution to provide this capability. It simply plugs into one of the two connectors on the back side of the LS Xenon Strobe or LS LED Strobe, providing precise manual control of the flash rate for the user. The LS-FCM be installed at any time.
Max Flash Rate is 120,000 fpm (2000 Hz) when used with LED Strobes and 12,500 fpm (208 Hz) with Xenon Strobes.
LS-FCM Operating Instructions
This control unit is designed for use with the LS Strobe series. With this control unit you can manually adjust the flash frequency of the stroboscope.
Warning: Please read all warnings as detailed in the instruction manual of the stroboscope.
How to use
1. Plug the LS-FCM into either one of the (2) jacks on the backside of the Strobe and tighten the screw.
2. To adjust the frequency, rotate the knob CCW (Counter Clockwise, left) to DECREASE flash frequency or CW (Clockwise, right) to INCREASE flash frequency.
(The rate of change will increase as you rotate the knob faster)